In measuring the microbial load in the air, it is possible to select the amount of air to be analyzed with Hytest Air, which is more advantageous and more accurate than the conventional method of petri opening.
Hytest Air Usage Areas
The microbial load of air can be measured with Hytest Air in all areas where airborne microbial load is important, such as production, packaging and storage areas, communal life areas.
Hytest Air Use
The device is switched on with the On / Off button.
With the Volume button, the amount of air to be sampled (10, 50, 100, 250, 500, 1000 Liters) is selected.
Open the lid of the medium, place the petri dish containing the medium, close the lid of the device.
Press the start button. The device stops when air absorption is finished.
Open the lid of the medium, remove the petri dish. The Petri dish is closed and placed in the incubator.
Press the off key to turn off the device.
Advantages of Hytest Air
It has ergonomic design that can be carried easily. LCD screen can display battery status and Hytest Air can be easily programmed with sliding menu system.
The desired amount of air can be selected as 10, 50, 100, 250, 500 and 1000 liters. When the set amount of air absorption ends, it gives an audible warning and ends the absorption.
Hytest Air can be operated by standing stationary in the environment sampled or by running it independently of the charge.
It charged with DC12V / 1A charging daptor. It can also be operated during charging.
For detailed information about Hytest Air, please contact us.